Study of output difference of two different ionization chambers for large fields used in radiotherapy
Many radiotherapy centers donâ€™t pay attention to effect of ionization champers type on accuracy of quality control measurements. They use any available ionization champers in all quality control and data entry measurements (1,2) . Many studies were carried out in this field to compare different ionization champers in small fields but large fields were not completely compared before (3.4). The aim of this work is to compare output factor in large field using two different sizes ionization champers connected to electrometer. Final out put were obtained from the farmer and smidflex dosimeter irradiated with 6 MV photon beams. Important field side ranging from 20 cm to 70 cm side field is measured in whole body radiation (5, 6).
For all examined large field sizes a difference ranging from 1% to 5 % was found when added to other calibration errors it will exceeds the acceptable margin. The largest difference was found in field side 70 cm this may be due to large scattering radiation
2- Olding T, Holmes O, Dejean P, McAuley KB, Nkongchu K, Santyr G, Schreiner LJ (2011) Small field dose deliveryevalua_ons using cone beam op_cal computed tomography-based polymer gel dosimetry. J Med Phys,36:3-14.
3- Van Dyk J. The modern Technology of Radiation Oncology. Madison: Medical Physics Publishing; 1999. p. 641-62.
4- Vrtar M. Fizikalni aspekti tehnike ozraÃ¨enja cijelog tijela (TBI). (Physical aspects of total body irradiation technique (TBI). Zbornik predavanja druÂ¹tvaza mernu tehniku Srbije - Sekcija za merenja u radioterapiji. Belgrade: 1988. p. 209-30.
5- INTERNATIONAL ATOMIC ENERGY AGENCY (IAEA), â€œCalibration of Dosimeters used in Radiotherapy: A manual sponsored by the IAEA and WHOâ€, IAEA Technical Report Series, TRS-374, IAEA, Vienna, Austria (1994).
6- INTERNATIONAL ATOMIC ENERGY AGENCY (IAEA), â€œThe use of plane parallel ionisation chambers in high energy electron and photon beamsâ€, IAEA Technical Report Series, TRS-381, IAEA, Vienna, Austria (1997).
7- INTERNATIONAL ATOMIC ENERGY AGENCY (IAEA), â€œAbsorbed dose determination in external beam radiotherapy: An international code of practice for dosimetry based on standards of absorbed dose to waterâ€, IAEA Technical Report Series, TRS-398, IAEA, Vienna, Austria (2000)
8- INTERNATIONAL ORGANIZATION FOR STANDARDIZATION (ISO), â€œGuide to expression of uncertainty in measurementâ€, ISO, Geneva, Switzerland (1992).
9- KHAN, F.M., â€œThe physics of radiation therapyâ€, Williams and Wilkins, Baltimore, Mary-land, U.S.A. (1994).
10- KLEVENHAGEN, S.C., â€œPhysics and dosimetry of therapy electron beamsâ€, Medical Physics Publishing, Madison, Wisconsin, U.S.A. (1993).
11- VAN DYK, J. (editor), â€œModern technology of radiation oncology: A compendium for medical physicists and radiation oncologistsâ€, Medical Physics Publishing, Madison, Wisconsin, U.S.A. (1999). Forman, G. 2003. An extensive empirical study of feature selection metrics for text classification. J. Mach. Learn. Res. 3 (Mar. 2003), 1289-1305.
Copyright (c) 2017 JOURNAL OF ADVANCES IN PHYSICS
This work is licensed under a Creative Commons Attribution 4.0 International License.
Authors retain the copyright of their manuscripts, and all Open Access articles are distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided that the original work is properly cited.
The use of general descriptive names, trade names, trademarks, and so forth in this publication, even if not specifically identified, does not imply that these names are not protected by the relevant laws and regulations. The submitting author is responsible for securing any permissions needed for the reuse of copyrighted materials included in the manuscript.
While the advice and information in this journal are believed to be true and accurate on the date of its going to press, neither the authors, the editors, nor the publisher can accept any legal responsibility for any errors or omissions that may be made. The publisher makes no warranty, express or implied, with respect to the material contained herein.